| Article Access Statistics|
| Viewed||204 |
| Printed||1 |
| Emailed||0 |
| PDF Downloaded||44 |
| Comments ||[Add] |
Click on image for details.
|Year : 2020
: 23 | Issue : 2 | Page
|Comparison of continuous cardiac output monitoring derived from regional impedance cardiography with continuous thermodilution technique in cardiac surgical patients
G Bhavya1, PS Nagaraja1, Naveen G Singh1, S Ragavendran1, N Sathish1, N Manjunath1, K Ashok Kumar2, Vinayak B Nayak1
1 Department of Anaesthesiology, Sri Jayadeva Institute of Cardiovascular Sciences and Research, Jayanagar, Bengaluru, Karnataka, India
2 Department of Cardiothoracic and Vascular Surgery, Sri Jayadeva Institute of Cardiovascular Sciences and Research, Jayanagar, Bengaluru, Karnataka, India
Click here for correspondence address and
|Date of Submission||01-Jan-2019|
|Date of Decision||27-Mar-2019|
|Date of Acceptance||23-Apr-2019|
|Date of Web Publication||07-Apr-2020|
| Abstract|| |
Background: Cardiac output (CO) assessment is a corner stone in advanced haemodynamic management, especially in critical ill patients. The present study was conducted to validate cardiac index and cardiac output by NICaS™ with the thermodilution technique using pulmonary artery catheter in post-operative cardiac surgical patients. Materials and Methods: This was a prospective observational clinical study conducted at a tertiary care hospital. 23 adult patients in the age range of 18-65 years who had undergone for elective coronary artery bypass grafting were included in the study. Results: Spearman's correlation coefficient of cardiac index between continuous Thermodilution (cTD) and Non-Invasive Cardiac System (NICaS™) showed a good correlation (r = 0.765, 95% confidence interval 0.70 to 0.82, P < 0.0001). There was a good correlation between cTD and NICaS™ for cardiac output (r = 0.759, 95% confidence interval 0.69 to 0.81, P < 0.0001), Bland-Altman plot for cardiac index between cTD and NICaS™ showed a mean bias of −0.66 ± 0.6919 with limits of agreement being −2.02 to 0.6936. Bland-Altman plot for cardiac output between cTD and NICaS™ showed a mean bias of −1.0386 ± 1.17 with limits of agreement being −3.34 to + 1.26. Percentage error for cardiac index and cardiac output were 64.78% and 64% respectively. Polar plot analysis showed an angular bias of 6.32° with radial limits of agreement being −8.114° to 20.75° for cardiac index and angular bias of 5.6682° with radial limits of agreement being −9.1422° to 20.4784° for cardiac output. Conclusion: NICaS™ demonstrated a good trending ability for both CI and CO. However, NICaS™ derived parameters are not interchangeable with the values derived from continuous thermodilution technique.
Keywords: Cardiac index, cardiac output, continuous thermodilution technique, NICaS™
|How to cite this article:|
Bhavya G, Nagaraja P S, Singh NG, Ragavendran S, Sathish N, Manjunath N, Kumar K A, Nayak VB. Comparison of continuous cardiac output monitoring derived from regional impedance cardiography with continuous thermodilution technique in cardiac surgical patients. Ann Card Anaesth 2020;23:189-92
|How to cite this URL:|
Bhavya G, Nagaraja P S, Singh NG, Ragavendran S, Sathish N, Manjunath N, Kumar K A, Nayak VB. Comparison of continuous cardiac output monitoring derived from regional impedance cardiography with continuous thermodilution technique in cardiac surgical patients. Ann Card Anaesth [serial online] 2020 [cited 2020 Jun 6];23:189-92. Available from: http://www.annals.in/text.asp?2020/23/2/189/282053
| Introduction|| |
Cardiac output (CO) assessment is a corner stone in advanced haemodynamic management, especially in critical ill patients. Currently, the average perioperative mortality after cardiac surgery is 1-2%. However, the incidence of cardiovascular morbidity remains high., Low cardiac-output syndrome (LCOS) is the most common and devastating complication which is characterized by reduced oxygen delivery (DO2) and subsequent tissue hypoxia.,,
Decrease in oxygen delivery results in anaerobic metabolism leading to hyperlactatemia, which is associated with increased post-operative mortality, morbidity and hospital length of stay. Hence, monitoring CO is vital in the early detection of an imbalance between oxygen demand and oxygen delivery.
The accurate measurement of CO is obtained by gold standard intermittent thermodilution technique through a pulmonary artery catheter (PAC). The continuous thermodilution technique (cTD) has an advantage over intermittent thermodilution technique as it displays continuous measurement of cardiac output which allows the clinician to monitor the trend. Since both are invasive in nature and the indications of floating a PAC is weighed against benefits versus risks, it is often less utilized.
Due to this limitation, there is an increased research on surrogate markers of CO in terms of oxygen delivery and oxygen consumption such as mixed venous oxygen saturation, arterial lactate levels and urine output.
In recent times, there has been burgeoning interest on non-invasive CO monitors like bioimpedance, bioreactance, applanation tonometry, partial carbon dioxide (CO2) rebreathing, pulse wave transit time and ultrasonic methods. There have been varied results among these non-invasive CO monitors.
Non-Invasive Cardiac System (NICaS™) is a new hemodynamic navigator using the principle of bioimpedance for non-invasive measurement of CO and its derivatives. To measure CO, an alternating electrical current (1.4 mA, 30 kHz) is passed through the patient via two pairs of tetrapolar electrodes (NI Medical, Hod Hasharon, Israel). One pair of electrodes is placed on the wrist over the radial artery, and the other pair is placed on the contralateral ankle over the posterior tibial artery.,
There is limited literature on NICaS™ derived hemodynamic parameters and its validation with invasive techniques in cardiac surgery.,,
Hence, the present study was conducted to validate cardiac index and cardiac output by NICaS™ with continuous thermodilution technique using pulmonary artery catheter in post-operative cardiac surgical patients.
| Materials and Methods|| |
This was a prospective observational clinical study conducted at a tertiary care hospital. After obtaining institutional ethical committee approval, the present study was conducted in the immediate post-operative patients.
Adult patients in the post-operative cardiac surgical unit—aged between 18 and 65 years—who had undergone elective coronary artery bypass grafting (CABG) were included in the study.
On the other hand, patients with post-operative myocardial infarction, valvular heart disease, emergency surgery, Bentall procedure, intracardiac shunts, post-operative arrhythmias, patients on Intra aortic balloon pulsation (IABP) and heart failure patients were excluded from the study.
Patients who were enrolled in the study were connected to both continuous cardiac output monitors - Continuous thermodilution technique (cTD) (Vigilance II™, Edwards Lifesciences, Irwin, USA) [Figure 1](a)] and Non-Invasive Cardiac System (NICaS™, NI medical, Petach Tikva, Israel) [Figure 1](b). Cardiac index (CI) and cardiac output (CO) measurements were obtained simultaneously at various time intervals until the patients were weaned from mechanical ventilation.
|Figure 1: (a) Continuous cardiac output monitoring by Vigilance II™ monitor. (b) Continuous cardiac output monitoring by NICaS™|
Click here to view
Hemodynamic monitoring by cTD was performed by placing pulmonary artery catheter (PAC) through right internal jugular vein whereas hemodynamic monitoring by NICaS™ was done by placing dual impedance electrodes on 2 limbs [one on volar aspect of the wrist and the other on the contralateral wrist/ankle] [Figure 2].
The results were presented as mean ± standard deviation. CI and CO values were analysed using Spearman's correlation to determine the strength of relationship between cTD and NICaS™. Correlation coefficient values range from being negatively correlated (-1) to uncorrelated (0) to positively correlated (+1) (0.0 is no association, +0.2 is weakly positive, +0.5 is moderately positive, +0.8 is strongly positive, +1.0 is perfectly positive).
Linear regression analysis was used to create a graphic representation of the relationship with the formula of the “best fit” line allowing the CI and CO measurements of NICaS™ to be calculated from cTD. The coefficient of determination (R2) is the proportion of variation in the dependent variable (NICaS™) can be explained by a linear regression model using the independent variable (cTD).
Bland-Altman limits of agreement (LOA) plots were constructed for these data. LOA plots visually represent the bias (mean difference) and variability (95% LOA) between two methods of measurement. 95% LOA were determined by 1.96*Standard Deviation (SD) of the mean difference of CI and CO values between cTD and NICaS™. Polar plot was also been constructed to know trending ability between the two monitors.
A P value <0.05 was considered statistically significant. Statistical analysis was performed using MedCalc version 12.2.1.
| Results|| |
A total of 23 patients were enrolled in the study, from whom 197 data sets were been analysed. Spearman's correlation coefficient of cardiac index between cTD and NICaS™ showed a strongly positive correlation (r = 0.765, 95% confidence interval 0.70 to 0.82, P < 0.0001). A strongly positive correlation was also observed between cTD and NICaS™ for cardiac output. (r = 0.759, 95% confidence interval 0.69 to 0.81, P < 0.0001).
Linear regression equations for CI and CO between cTD and NICaS™ were:
y = −0.39 + 1.49× (R2 = 0.65, P < 0.0001) and y = −0.87 + 1.52× (R2 = 0.59, P < 0.0001) [y- NICaS™; × - cTD], respectively.
Bland-Altman plot for CI between cTD and NICaS™ showed a bias of − 0.66 ± 0.6919 with LOA being −2.02 to 0.6936 [Figure 3]. Whereas for CO bias was −1.0386 ± 1.17 with LOA: −3.34 to +1.26 [Figure 4].
|Figure 3: Bland-Altman plot to compare CI between cTD (Vigilance II™) and NICaS™|
Click here to view
|Figure 4: Bland-Altman plot to compare CO between cTD (Vigilance II™) and NICaS™|
Click here to view
Percentage error for cardiac index and cardiac output were 64.78% and 64% respectively. Polar plot analysis showed an angular bias of 6.32° with radial limits of agreement being −8.114° to 20.75° for cardiac index [Figure 5] and angular bias of 5.6682° with radial limits of agreement being −9.1422° to 20.4784° for cardiac output.
| Discussion|| |
The NICaS™ apparatus measures the bioimpedance of the systemic circulation and calculates the stroke volume (SV) using the Tsoglin and Frinerman formula: SV = △R/R × ρ × L2/Ri × (α + β)/β × Kw× HF where △ R is the change in impedance during the cardiac cycle (Ω), R is the basal resistance (Ω), ρ is the blood electrical resistivity, L is the patient's height, and Ri is the corrected basal resistance according to gender and age. KW is a correcting factor for body weight, HF is a hydration factor related to the body water composition, α + β is equal to the electrocardiogram R-R interval, and β is the diastolic time interval.,,
The present study demonstrates a strong positive correlation for CI and CO with a significant coefficient of determination (R2) for CI and CO being 0.65 and 0.59, respectively, with a P value <0.0001 which shows a good trending ability between cTD and NICaS™ monitors.
The bias for CI and CO were −0.66 ± 0.6919 and −1.0386 ± 1.17, respectively, but with wide LOA and an increased percentage error of 64% between two monitors signifying less precision.
Anat Lavie et al. used the NICaS™ system to compare the hemodynamic parameters of 17 women with severe preeclampsia to a cohort of healthy normotensive pregnant women with a singleton pregnancy at term. The NICaS™ device noninvasively demonstrated low CO and high total peripheral resistance profiles in the preeclampsia group compared to control. They concluded that the NICaS™ device may help the clinician to customize the most optimal management for individual parturients.
Matsuda et al. evaluated NICaS™ derived CO values in 8 preoperative patients with pheochromocytoma and compared with simultaneous central blood volume (CBV) values measured by a conventional indicator dilution method using131 I - labeled human serum albumin. The NICaS™ -derived CO values significantly correlated with CBV values measured by131 I-labeled human serum albumin (4.86 ± 1.05 L/min vs 4.79 ± 1.02 L/min; r = 0.906; P = 0.002). Sequential NICaS™ derived CO values confirmed that CBV increased after preoperative treatment with an α-blocker—with or without volume loading. The results of this study indicated that NICaS™ can be used to accurately and non-invasively evaluate the hemodynamic status. By sequential monitoring of NICaS™ -derived CO values, the authors could confirm if adequate CBV in a patient with pheochromocytoma is obtained by preoperative medical treatment with α-blockers or volume loading, to avoid perioperative complications.
Michael J Germain et al. evaluated stroke volume measurements using bioimpedance cardiography and doppler echocardiography in 17 patients undergoing maintainance hemodialysis. The authors concluded that NICaS™ Stroke volume (SV) measurements are similar to and strongly correlated with echocardiographic SV measurements.
B W Allwood et al. compared whole body impedance cardiography with thermodilution and modified Fick method of CO in 14 patients with pulmonary hypertension undergoing right heart catheterization. The authors concluded that whole body impedance cardiography may provide accurate measurements of cardiac output in patients with pulmonary hypertension and could potentially be a tool for assessing response to therapy.
Cotter MD et al. enrolled 122 patients in three different groups during cardiac catheterization (n = 40) before, during, and after coronary artery bypass surgery (n = 51); and while being treated for acute congestive heart failure (CHF) (n = 31). 418 paired CO measurements were obtained. The overall correlation between the NICaS™ cardiac index (CI) and the thermodilution CI was r = 0.886, with a small bias (0.0009 ± 0.684 L) [mean ± 2 SD], and this finding was consistent within each group of patients.
Though the studies conducted by Anat Lavie et al., Matsuda et al., Allwood BW et al. and Cotter et al. showed good accuracy and precision using NICaS™ for guiding therapeutic intervention which is not in agreement with the present study. Critchley et al. has demonstrated that polar plot analysis showing an angular bias of <5° and radial limits of agreement ±30° have good trending ability of the monitor. In the present study, polar plot analysis showed an angular bias of 6.32° with radial limits of agreement being −8.114° to 20.75° for cardiac index and angular bias of 5.6682° with radial limits of agreement being −9.1422° to 20.4784° for cardiac output.
| Conclusion|| |
NICaS™ demonstrated a good trending ability for both CI and CO. However, NICaS™ derived parameters are not interchangeable with the values derived from continuous thermodilution technique.
The authors thank NICaS™, NI medical, Petach Tikva,Israel (Sandor Medicaids Pvt. Ltd., India) for providing the technical equipment (NICaS™) needed for this study.
Financial support and sponsorship
Conflicts of interest
This manuscript had been presented with Janak Mehta Award in IACTA 2019.
| References|| |
Bridgewater B. Mortality data in adult cardiac surgery for named surgeons: Retrospective examination of prospectively collected data on coronary artery surgery and aortic valve replacement. BMJ 2005;330:506-10.
Chen JC, Kaul P, Levy JH, Haverich A, Menasche P, Smith PK, et al
. Myocardial infarction following coronary artery bypass graft surgery increases health care resource utilization. Crit Care Med 2007;35:1296-301.
Maganti MD, Rao V, Borger MA, Ivanov J, David TE. Predictors of low cardiac output syndrome after isolated aortic valve surgery. Circulation 2005;112:I448-52.
Maganti M, Badiwala M, Sheikh A, Scully H, Feindel C, David TE, et al
. Predictors of low cardiac output syndrome after isolated mitral valve surgery. J Thorac Cardiovasc Surg 2010;140:790-6.
Vincent JL, De Backer D. Circulatory shock. N
Engl J Med 2013;369:1726-34.
Lavie A, Ram M, Lev S, Blecher Y, Amikam U, Shulman Y, et al
. Maternal cardiovascular hemodynamics in normotensive versus preeclamptic pregnancies: A prospective longitudinal study using a noninvasive cardiac system (NICaS™). BMC Pregnancy Childbirth 2018;18:229.
Matsuda Y, Kawate H, Shimada S, Matsuzaki C, Nagata H, Adachi M. Perioperative sequential monitoring of hemodynamic parameters in patients with pheochromocytoma using the non-invasive cardiac system (NICaS). Endocr J 2014;61:571-5.
Germain MJ, Joubert J, O'Grady D, Nathanson BH, Chait Y, Levin NW. Comparison of stroke volume measurements during hemodialysis using bioimpedance cardiography and echocardiography. Hemodial Int 2018;22:201-8.
Allwood BW, Witkin AS, Rodriguez-Lopez JM, Channick RN. Comparison of whole body impedance cardiography with thermodilution and modified fick measures of cardiac output, in patients with pulmonary hypertension. Am J Respir Crit Care Med 2018;197:A4377.
Cotter G, Moshkovitz Y, Kaluski E, Cohen AJ, Miller H, Goor D, et al
. Accurate, noninvasive continuous monitoring of cardiac output by whole-body electrical bioimpedence. Chest 2004;125:1431-40.
Critchley LA, Lee A, Ho AM. A critical review of the ability of continuous cardiac output monitors to measure trends in cardiac output. Anaesth Analg 2010;111:1180-92.
Naveen G Singh
Department of Anaesthesia, Sri Jayadeva Institute of Cardiovascular Sciences and Research, Bangalore - 560 069, Karnataka
Source of Support: None, Conflict of Interest: None
[Figure 1], [Figure 2], [Figure 3], [Figure 4], [Figure 5]